gas chromatography · pdf file 1.packed column: columns are available in a packed manner s.p...
Click here to load reader
Post on 24-Jul-2020
Embed Size (px)
Gas Chromatography Presented By -
Dr. Sohail Ayub Department of Civil Engineering AMU Aligarh India Email – firstname.lastname@example.org
What is Gas Chromatography?
• It is also known as… – Gas-Liquid Chromatography (GLC)
Separation of gaseous & volatile substances Simple & efficient in regard to separation
GC consists of GSC (gas solid chromatography) GLC (gas liquid chromatography
Gas → M.P Solid / Liquid → S.P GSC not used because of limited no. of S.P GSC principle is ADSORPTION GLC principle is PARTITION
Sample to be separated is converted into vapour And mixed with gaseous M.P Component more soluble in the S.P → travels slower Component less soluble in the S.P → travels faster Components are separated according to their
Partition Co-efficient Criteria for compounds to be analyzed by G.C 1.VOLATILITY: 2.THERMOSTABILITY:
What is Gas Chromatography?
• The father of modern gas chromatography is Nobel Prize winner John Porter Martin, who also developed the first liquid-gas chromatograph. (1950)
The Next Generation in Gas Chromatography
How a Gas Chromatography Machine Works
– First, a vaporized sample is injected onto the chromatographic column.
– Second, the sample moves through the column through the flow of inert gas.
– Third, the components are recorded as a sequence of peaks as they leave the column.
– Deals with both the stationary phase and
the mobile phase. • Mobile – inert gas used as carrier. • Stationary – liquid coated on a solid or a solid
within a column.
Chromatographic Separation • Chromatographic Separation
– In the mobile phase, components of the sample are uniquely drawn to the stationary phase and thus, enter this phase at different times.
– The parts of the sample are separated within the column.
– Compounds used at the stationary phase reach the detector at unique times and produce a series of peaks along a time sequence.
Chromatographic Separation (continued) – The peaks can then be read and analyzed by a
forensic scientist to determine the exact components of the mixture.
– Retention time is determined by each component reaching the detector at a characteristic time.
– The number of components in a sample is determined by the number of peaks.
– The amount of a given component in a sample is determined by the area under the peaks.
– The identity of components can be determined by the given retention times.
Peaks and Data
• Carrier gas • Flow regulators & Flow meters • Injection devices • Columns • Temperature control devices • Detectors • Recorders & Integrators
CARRIER GAS » Hydrogen better thermal conductivity disadvantage: it reacts with unsaturated
compounds & inflammable » Helium excellent thermal conductivity it is expensive » Nitrogen reduced sensitivity it is inexpensive
Requirements of a carrier gas
Inertness Suitable for the detector High purity Easily available Cheap Should not cause the risk of fire Should give best column performance
Flow regulators & Flow meters deliver the gas with uniform pressure/flow rate flow meters:- Rota meter & Soap bubble
flow meter Rota meter placed before column inlet it has a glass tube with a float held on to a
spring. the level of the float is determined by the
flow rate of carrier gas
Soap Bubble Meter ◊ Similar to Rota meter & instead of a float,
soap bubble formed indicates the flow rate
Gases can be introduced into the column by valve devices
liquids can be injected through loop or septum devices
COLUMNS • Important part of GC • Made up of glass or stainless steel • Glass column- inert , highly fragile
COLUMNS can be classified Depending on its use 1. Analytical column 1-1.5 meters length & 3-6 mm d.m 2. Preparative column 3-6 meters length, 6-9mm d.m
Depending on its nature 1.Packed column: columns are available in
a packed manner S.P for GLC: polyethylene glycol, esters,
amides, hydrocarbons, polysiloxanes… 2.Open tubular or Capillary column or
Golay column Long capillary tubing 30-90 M in length Uniform & narrow d.m of 0.025 - 0.075 cm Made up of stainless steel & form of a coil Disadvantage: more sample cannot loaded
3.SCOT columns (Support coated open tubular column Improved version of Golay / Capillary
columns, have small sample capacity Made by depositing a micron size
porous layer of supporting material on the inner wall of the capillary column Then coated with a thin film of liquid
Equilibration of the column Before introduction of the sample Column is attached to instrument &
desired flow rate by flow regulators Set desired temp. Conditioning is achieved by passing
carrier gas for 24 hours
Temperature Control Devices Preheaters: convert sample into its vapour
form, present along with injecting devices Thermostatically controlled oven: temperature maintenance in a column is
highly essential for efficient separation. Two types of operations Isothermal programming:- Linear programming:- this method is
efficient for separation of complex mixtures
• Isothermal • Gradient
0 10 20 30 40 50 60
Instrumentation - Oven
DETECTORS Heart of the apparatus The requirements of an ideal detector are- Applicability to wide range of samples Rapidity High sensitivity Linearity Response should be unaffected by
temperature, flow rate… Non destructive Simple & inexpensive
Measures the changes of thermal conductivity due to the sample (g). Sample can be recovered.
1.Thermal Conductivity Detector (Katharometer, Hot Wire Detector)
Thermal Conductivity Basics
When the carrier gas is contaminated by sample , the cooling effect of the gas changes. The difference in cooling is used to generate the detector signal.
The TCD is a nondestructive, concentration sensing detector. A heated filament is cooled by the flow of carrier gas.
When a separated compound elutes from the column , the thermal conductivity of the mixture of carrier gas and compound gas is lowered. The filament in the sample column becomes hotter than the control column.
The imbalance between control and sample filament temperature is measured by a simple gadget and a signal is recorded
Thermal Conductivity Detector
� Measures heat loss from a hot filament – � filament heated to const T • when only carrier gas flows heat loss to
metal block is constant, filament T remains constant.
• when an analyte species flows past the filament generally thermal conductivity goes
down, T of filament will rise. (resistance of the filament will rise).
Relative Thermal Conductivity
Compound Relative Thermal Conductivity Carbon Tetrachloride 0.05 Benzene 0.11 Hexane 0.12 Argon 0.12 Methanol 0.13 Nitrogen 0.17 Helium 1.00 Hydrogen 1.28
Advantages of Katharometer Linearity is good Applicable to most compounds Non destructive Simple & inexpensive Disadvantages Low sensitivity Affected by fluctuations in temperature and
flow rate Biological samples cannot be analyzed
Flame Ionization Detector Destructive detector The effluent from the column is mixed with H
& air, and ignited. Organic compounds burning in the flame
produce ions and electrons, which can conduct electricity through the flame.
A large electrical potential is applied at the burner tip
The ions collected on collector or electrode and were recorded on recorder due to electric current.
FIDs are mass sensitive rather than conc. sensitive
ADVANTAGES: • µg quantities of the solute can be
detected • Stable • Responds to most of the organic
compounds • Linearity is excellent • DA: destroy the sample